Application of Solid Phase Microextraction (SPME) Sampler for Determination of Carbon Disulfide in Air
Carbon disulfide is used predominantly in the manufacture. It has affects the nervous system. In this study, the applicability of SPME as a passive sampler for determination of carbon disulphide in air was studied. Effect of sampler and environmental parameters on uptake of Carbon disulphide was studied as well. Four fibers were tested to select the best sampler for determine carbon disulfide in ambient air. A standard generation chamber was built in the laboratory and was used to test the SPME. Analysis SPME samples were carried out by a gas chromatography-Mass spectrometry and results were compared with data obtained with National Institute for Occupational Safety and Health (NIOSH) method 1600. Polymethylsiloxane-carboxen (PDMS/CAR) showed the most effective stationary phase material for sorbing BTEX among other materials (polyacrylate, PDMS, PDMS/divinylbenzene). Its linearity range in exposed mode was less than 10 minutes but with its retracted mode application, its linearity increased up to 8 hours. Temperature had not linear effect on uptake of pollutant in temperatures lower than 25, it has positive effect and above this range it has negative effect. Relative humidity had negative effect on mass loaded on fiber. Velocity in range of static to 0.5 m/s had no significant effect. The precision of the method was 4.18% relative standard deviation (RSD). The detection limit for carbon disulfide in the GC/MS system in SIM mode was 6.7 ng per sample. SPME is a good alternative for sampling of carbon disulfide in air. However, for the situations in high humidity values it should be used with care.
OSHA: OccupationalSafety and Health Guideline for Carbon Disulfide. Available from: http://www.osha.gov/SLTC/healthguidelines/carbondisulfide/recognition.html
Beauchamp R, Bus JS, Popp JA, Boreiko CJ, Goldberg L. A critical review of the literature on carbon disulfide toxicity. Crit Rev Toxicol 1983;11(3): 169-278.
Kotseva K. Occupational exposure to low concentrations of carbon disulfide as a risk factor for hypercholesterolaemia. Int Arch Occ Env Hea 2000;74(1): 38-42.
Price B., Bergman TS, Rodríguez M, Henrich RT, Moran EJ.A review of carbon disulfide exposure data and the association between carbon disulfide exposure and ischemic heart disease mortality. Regul Toxicol and Pharm 1997;26(1): 119-128.
Kotseva K, Braeckman L, De Bacquer D, Bulat P, Vanhoorne M. Cardiovascular effects in viscose rayon workers exposed to carbon disulfide. Int J Occ Env Hea 2001;7(1): 7-13.
Lewis JG, Graham DG, Valentine WM, Morris RW, Morgan DL, Sills RC. Exposure of C57BL/6 mice to carbon disulfide induces early lesions of atherosclerosis and enhances arterial fatty deposits induced by a high fat diet. Toxico Sci 1999;49(1): 124-132.
Saillenfait A, Bonnet P, De Ceaurriz J.Effects of inhalation exposure to carbon disulfide and its combination with hydrogen sulfide on embryonal and fetal development in rats.Toxico Let 1989;48(1): 57-66.
American Conference of Governmental Industrial Hygienists. TLVs and BEIs: threshold limit values for chemical substances and physical agents; biological exposure indices. Cincinnati, Ohio: USA 2013.
National Institute for Occupational Safety and Health, (NIOSH). Method 1600 in NIOSH Manual of Analytical Methods. Eller PM, Cassinelli ME, eds. Cincinnati, Ohio: USA1994
Krupa S. Legge A.Passive sampling of ambient, gaseous air pollutants: an assessment from an ecological perspective. Environ Pollut 2000; 107(1): 31-45.
Zare Sakhvidi M J, Bahrami A, Afkhami A, Rafiei A. Development of diffusive solid phase microextraction method for sampling of epichlorohydrin in air. Int J Environ Anal Chem 2012; 92(12): 1365-1377.
Chai M, Pawliszyn J.Analysis of environmental air samples by solid-phase microextraction and gas chromatography/ion trap mass spectrometry. Environ Science Techno 1995;29(3): 693-701.
Zare Sakhvidi MJ, Bahrami AR, Ghiasvand A, Mahjub H, Tuduri L SPME-based air sampling method for inhalation exposure assessment studies: case study on perchlorethylene exposure in dry cleaning. Environ Monit Assess 2013 185(6) 4933-4941
Zare Sakhvidi MJ, Bahrami AR, Afkhami A, Rafiei A. Development of diffusive solid phase microextraction method for sampling of epichlorohydrin in air. Int J Environ Analyt Chem 2012; 92(12): 1365-1377.
Zare Sakhvidi MJ, Bahrami AR, Ghiasvand A, Mahjub H, Tuduri L. Field application of SPME as a novel tool for occupational exposure assessment with inhalational anesthetics. Environ Monit Assess 2012;184(11): 6483-6490.
Zare Sakhvidi M J, Bahrami AR, Ghiasvand A, Mahjub H, Tuduri L. Determination of inhalational anesthetics in field and laboratory by SPME GC/MS. Anal Lett 2012;45(4): 375–385.
Tuduri L, Desauziers V, Fanlo JL. A simple calibration procedure for volatile organic compounds sampling in air with adsorptive solid-phase microextraction fibres. Analyst 2003;128(8): 1028-1032.
Fuller EN, Schettler PD, Giddings JC. New methods for prediction of binary gas-phase diffusion coefficients. Ind Eng Chem 1966; 58 (5):18–27.
EURACHEM/CITAC Guide. The Fitness for Purpose of Analytical Methods. 2nded, Teddington, Middlesex, UK, 1998 Available from www.eurachem.org.
Ouyang G, Zhao W, Bragg L, Qin Z, Alaee M, Pawliszyn J. Time-weighted average water sampling in Lake Ontario with solid-phase microextraction passive samplers. Environ Sci Technol 2007;41(11): 4026-4031.
Seethapathy S, Górecki T, and Li X. Passive sampling in environmental analysis. J Chromatogr A 2008;1184(1): 234-253.
Copyright (c) 2015 International Journal of Occupational Hygiene
This work is licensed under a Creative Commons Attribution-NonCommercial 4.0 International License.